Electronic Data Collection and Analysis System
Yin Foo, Des; Foley, Margaret
E-Publications; PublicationNT; E-Books; Report ; 39/1992
Made available via the Publications (Legal Deposit) Act 2004 (NT).
Power and Water Authority
Report ; 39/1992
Check within Publication or with content Publisher.
Technical Report WRD92039 Viewed at 15:07:17 on 29/07/2010 Page 41 of 192. For example, the scatter of logged head from a test ",here the transducer was outside its conduit, at the pump inlet and with a free hanging pump column was not significantly different to that from another test where the transducer was set inside a perforated conduit, 4m above the pump inlet, and where the pump column was centralised. The effectiveness of various configurations used need to be n,-affirmed under conditions other than those available during this series of tests. The worst scatter encountered varied from the compared to the site (RN 20544) differed. order of 300mm at the 3m scatter found at the original test site HcNillans Road test where pump setting, rate and bore construction The only controllable factor found to influence the vibrations using a helical rotor type pump was the pump speed. separate series of tests The Nindata and Unidata This was examined in detail through a on bore RN 20544 (see Reference 18). (borrowed from W.A.lv.A.) systems were The Wesdata system (which at that conduit) could not be fitted downhole. used for these tests. time required a 50mm il. Hono 620 pump with 50mrn column, set at 40m below ground vlaS used for these tests. The tests were designed initially to determine whether the pump motor or shaft speed were variables associated with the "scatter" of data. If so, then a working range of speeds should be determined so that these could be specified to produce a scatter of data to within acceptable limits. In one test the pump shaft speed was kept const~e,nt at 3 different engine speeds (using the gearbox). In the test "HH'IEST1" the engine was run at three different speeds in each of four gears (Refer Figure 3.6). 188.8.131.52 pumped Bore Results (Electric Submersible P'.l11lps) Two systems, a Mindata and a Wesdata were trialled in a bore pumped by a "Multi-Motor" submersible pump. The logged
Aboriginal and Torres Strait Islander people are advised that this website may contain the names, voices and images of people who have died, as well as other culturally sensitive content. Please be aware that some collection items may use outdated phrases or words which reflect the attitude of the creator at the time, and are now considered offensive.
We use temporary cookies on this site to provide functionality.
You are welcome to provide further information or feedback about this item by emailing TerritoryStories@nt.gov.au